Weighing devices used for commercial or laboratory use are often designated within accuracy classes, which are usually referred to simply as classes. What do these classifications mean? Let's take a look!
We can't talk about classes without first discussing scale divisions. A scale division is the smallest subdivision or unit that can be indicated by a scale's digital display. It's a term commonly used interchangeably with readability or resolution.
Division can be determined by dividing the scale's capacity by its readability. If you have a scale with a 1000kg capacity and a 0.1kg readability (1000 divided by 0.1) has 10,000 divisions. In this example, if you have items that each weighed 0.1lb, you would be able to put 10,000 of them on the scale before you reached the maximum capacity of that scale.
Discussion of scale divisions often includes reference to "d" and "e" values. The "d" refers to the value of the scale division, whilst the more precise "e" refers to the verification scale division. Used mainly for commercial transactions, "e" is the smallest approved scale increment that can be used to determine a selling price by weight.
Accuracy Class Chart from OIML R 76-1
The class definitions remain the same globally (though some countries may refer to Class I, for example, as Special class, as noted in the headings below). Universal definitions like this can promote collaboration, as well as simplify scientific and commercial applications. Click here for a handy infographic.
Class I scales, which are ideal for precision lab work, have a minimum of 50,000 divisions. They display results in increments of 1mg or better. Semi-micro and analytical balances are often rated Class I.
Class II scales have up to 100,000 divisions. Typical uses include laboratory weighing, weighing of precious metals and gems, and grain test scales.
Our Highland portable precision balance is an example of a Class II balance.
Class III scales are designed for industrial or commercial use, including tasks like animal weighing, food weighing, postal use, scales used to determine laundry charges and vehicle on-board weighing systems (with capacities up to 30,000 lbs.). They have a maximum of 10,000 divisions.
Class III L
Like Class III scales, Class III L are also used for commercial use, but they offer a much larger capacity and are used for weighing vehicles, axle loads, livestock, railway tracks, vehicle on-board weighing systems (with capacities greater than 30,000 lbs.), cranes and hoppers (other than grain). Like the standard Class III scales, they also have a maximum of 10,000 divisions.
Class IIII – wheel-load weighers and portable axle-load weighers used for highway weight enforcement –have up to 1,000 divisions.
How Do Accuracy Classes Relate to Trade Approval?
Whether NTEP, Measurement Canada, CE, OIML, NMI or NRCS, trade approvals rely on the "e" verification scale division.
Whilst not all weighing devices must be legal-for-trade, the trade approvals do use accuracy class standards as part of their approval process.
If you have any questions about how Adam products are classified or you need a weighing device in a certain class, ask us!